Medical Mini Oxygen Regulator, Hospital Equipment Medical Equipment
|FOB Price:||US $10-14 / Piece|
|Min. Order:||100 Pieces|
|Min. Order||FOB Price|
|100 Pieces||US $10-14/ Piece|
|Payment Terms:||L/C, T/T, Western Union, Paypal|
- Model NO.: DY-C3
- Customized: Customized
- Type: Oxygen Regulator
- Input Pressure: 150bar
- Safety Valve: 3.5bar±0.5bar
- Material: Brass
- Function: Oxygen Supply
- Trademark: Dengyue
- Origin: Zhejiang
- Medical Device Regulatory Type: Type 1
- Certification: CE, ISO13485
- Gas Type: Oxygen
- Output Pressure: 2bar-3bar
- Flowmeter Range: 1L/Min-10L/15L/Min
- Connecting Thread: Male G5/8
- Group: All
- Transport Package: Paper Box
- HS Code: 901920000
China Hospital Use Oxygen Regulator Medical Equipment Supplier
|Automatic Discharge Pressure of The Safety Valve:||0.35±0.05Mpa|
|The Range of Flow:||1-15L/min|
The flowmeter with humidifier is used in reducing gas pressure and adjusting the gas volume, which rescue patient and treat as oxygen therapy by hospital.Also may make using of other gas reduced pressure and the control flow.
Product main feature:
The product has artistic outlook, and flow control comfortable and stable.
The reductor is adjustable;The structure is unique, and the performance is reliable.
The stainless steel floater and the orange substrate flowmeter have a especially visual effect;
The international standard connect way, is suitable for all kinds of gas bottle installation.
Range of application
Give oxygen for first-aid or oxygen uptake for the people who are lack of oxygen.
How to use
1.Open the valve of oxygen cylinder after connect the screw nut of air intake closely.
2.Connect the oxygen pipe to the oxygen outlet firmly.
3.Demount the humidifying cup(anti clock wise rotation),put some distilled water in (purified water or plain boiled water can also take place),the water quantity should be between the bottom and top tick mark,then tighten up the bottle cap to the humidifying cup(clockwise rotation).
4.Rotate the flow regulating valve to the rate you need (anti clockwise to up,clockwise to down,till to close),then put it to nose to use.
5.Close the valve of oxygen cylinder firstly while not using,then close the flow regulating valve until the pointer of oxygen manometer return to zero.
1.Make sure there is no ash on the connector and check the screw thread,then install the inhalator.
2.Make sure the flow meter is vertical to the floor and flow regulating valve is closed,then open the valve of oxygen cylinder slowly,regulate the flow after the high pressure gauge shows the pressure of the oxygen cylinder.
3.No drastic shock,far away with the corrosive substance,and no oil stain.
4.The set screw of the safety valve can't be rotated at random,non-professional people are forbidden to dismount.
Condition for storage
Should be kept in the environment whose humidity is less than 90%,no corrosive gas and drafty.
Quality guarantee period
We will fix or change the oxygen inhalator within a half and a year after the day of living the company when it can't proper functioning,it should be under the condition of obeying the rule of storage and using.
1.Forbid to use the Buoy Type Oxygen Regulator to take oxygen atomization treatment.
2.Forbid to dismount by yourself.
3.The defects liability period is 18 months.
Yuyao dengyue medical equipment Co.,Ltd is involved in the manufacturing of medical equipment since 1987. we are a local company with a team of knowledgeable people with expertise in the medical.the company specializes in the manufacturing of medical gas regulator,nebulizer,wall medical flow meter,gas outlet,cylinder valve and at the same time we carry stock medical gas cylinders with other related product.The company in 2012, had passed CE certification and ISO 13485 quality management system certification.We have exported our product to Philippines,Vietnam,Middle East,East and South Asia and so on.At the same time,we have received encouraging feedbacks from both international and domestic markets.