Aneroid Sphygmomanometer with Single Head Stethoscope, Sphygmomanometer
|FOB Unit Price:||US $5.5 US $5|
|Purchase Qty. (Pieces)||FOB Unit Price|
|Transport Package:||Paper Box|
|Payment Terms:||L/C, T/T, Paypal, Money Gram, Western Union|
- Model NO.: XYB04
- Type: Blood Pressure Meter
- Group: Middle-aged and Old
- Cuff Color: Blue, Black etc.
- Material of Bladder: PVC or Latex
- Bussiness Type: Manufacturer
- Specification: CE ISO13485
- HS Code: 9018902000
- Classification: Physiological Functions of Diagnosis and Monitoring Equipment
- Certification: CE, ISO13485
- Style: Blood Pressure Monitor
- Material of Bulb: PVC or Latex
- Certificatio: ISO, Ce
- Trademark: Dengyue
- Origin: Zhejiang
Read Aneroid Sphygmomanometer with latex bladder
1.OEM is available
3.Sample is available
Aneroid Sphygmomanometer with stethoscope
0-300mmHg or 20-300mmHg
PVC or Latex
PVC or Latex
A.The product is in compliance with EN1060Non-invasive
Practice the following procedure several times to become
Familiar with the equipment Before taking your blood to relax
And rest for at least 15minutes.This will reduce the error due
To physical activity.If you re-right handed.Slip the pressure
Cuff over your left arm and extent your left arm palm up.
Whether sitting or lying,be sure that your arm rests at the
Same level as your heart.Turn the cuff so that the stethoscope
Diaphragm (sound head) is on the inside of the arm just above
The elbow.This locates it over the large artery near the hollow
Of the elbow.(this artery can be also found by feeling for the
Pulse of the artery ) Grasp the end of the cuff and pull it snug.
Then wrap it around and over your arm.Pressing together with
Velcro tape to bind the cuff securely.
2.Inflate the cuff
Feel the pulse of the artery with your fingertips to be sure the
Heads of the stethoscope is located directly over it.Insert the
Earpieces of the stethoscope (adjust by turning earpieces to fit
Comfortably in your ears ) Hold the gauge in your left hand or
Clip onto the chug on the loop provided for reading.Close the
Air-flow valve on the bulb (turn clockwise) DO NOT
OVERTIGHTEN.Inflate the cuff by repeatedly squeezing the
Bulb with your right hand.Listen to the pulse beat whilst
Inflating the cuff.Watch the gauge.When you can no longer
Hear the pulse beat,raise the pressure an additional 30mm.
3.Slowly deflate the cuff
Slowly open the air-flow valve by turning counter clockwise so
That the pressure drops 2-4mmHg with each beat of your heart.
(This will usually mean a drop of one to two marks on the gauge
The rate of deflation is important for accurate reading
Remember, high pressure of the cuff has shut of all blood.
Flow to the arm.So do not leave it inflated fully any longer than
After opening the air-flow valve,listen carefully for a pulse
Beat.The moment you hear the faint rhythmic tapping or
Thumping sound of the pulse beat, note tbe reading on the
Gauge.This is your systolic(upper)blood pressure.
It may be different at first for you to detect the sounds of the
Pulse beat.Listen carefully and acquaint yourself with these
Unfamiliar sounds.You should also notice the needle on the
Gauge "bounce" when your systolic pressure is reached.
Allow the pressure to continue dropping at the same rate as
Before 2-4mmHg per second Listen carefully with the
Stethoscope The sounds you hear will change the phases.
From the first sharp tagging or thud.They will soften to blowing
Or swishing sounds.Watch the falling needle.At the exact
Point when you can nolonger hear the sounds, read the gauge,
This is your diastolic (lower) blood pressure reading.
With proper care.Your blood pressure Monitor will provide
Many years ofreliable service To insure a well functioning unit,
Follow these basic rules.
*Pressure gauge should be handled with care,do not drop or
Jar Needle should indicate zero when cuff is fully deflated.
*Always deflate cuff before storing.
*Avoid bitting objects against surface of stethoscope.
Under normal condition, the device is recommended to be
Calibrated every two years.Calibration should also be done
After repair.Calibration should be carried out at point 50mmHg